The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Published (Last):||5 July 2009|
|PDF File Size:||3.78 Mb|
|ePub File Size:||3.65 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fluorescence Tests Required Materials BioTek offers a liquid test kit PN containing the microplates and solutions used in this procedure. Specifications BioTek Instruments, Inc.
Installation BioTek Instruments, Inc. By pressing ‘print’ button you will print only current page. Connect the Host Computer Page 48 – Page Appendix D: Rinse The Fluid Lines Page 97 H12 position is 1.
Plug the lamp cables into the power source located to the manuzl of the lamp. Check the microplate carrier to ensure it is clear of debris.
Depress tab to separate the connectors Remove the thumbscrew located in the left rear of the instrument and set it aside. Tip Prime before dispensing. Getting Started Inside the Synergy HT, two Teflon tubes transport fluid from the tubing ports on the rear of the instrument to the bt injectors.
Absorbance Plate Test Cable clamp Locate the injector heads.
Synergt the fittings; no leaks should be detected. Hazards And Precautions If decontaminating with alcoholremove the trough and plate and let them air- dry.
Page Appendix B: Home Sensor Verification Errors The Synergy HT is attached to a shipping panel that has two handles for lifting. Dilute using the 0. Status String Format Do not touch the lens with your fingers!
BioTek Synergy HT Operator’s Manual
Running A Dispense Protocol optional Page 56 Chapter 2: The calibration sticker is affixed directly to the plate. Top optical probe Screw for lowering and raising the top optical probe Incubator Top probe hanger, with housing two shoulder screws Figure Procedure For Models Without Injectors Cleaning The Optical Probes cover.
Page 40 Lift out the module and place it on a level surface. Product Registration Chapter 2: Run The Absorbance Plate Test Use new microplates from sealed packages.
When cleaning the absorbance lens with the swab, apply very little pressure to the lens!
To do this, depress the small tab pictured below and separate the connectors. Controlling The Dispense Module synerby Gen5 protocol in advance as well, to ensure that the intended reading parameters are used and to avoid any last-minute corrections. Be sure to seat the injector tips securely when reinstalling.
Exposed surfaces may be cleaned not decontaminated with a cloth moistened not soaked with water or water and a mild detergent. Page 62 Clean both sides of the filter with lens paper. Register With Biotek Use a clean, dry cloth to dry all wet surfaces on the instrument and the Dispense module.
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
This error indicates significant variations in background electronic noise were detected when blocking the light and increasing the gain to maximum. The outlet tubes are opaque PTFE tubes with threaded fittings on each end that are used to deliver fluid from the syringes to the instrument.
Required bioek outside when locked in If the carrier is inside the read chamber and the probe needs to move down but the door is locked, the carrier cannot move out of the way of the top probe assembly. Please contact BioTek if you have misplaced either of these items. Verify that the Test Plate is within its calibration certification period.