Outlines procedures required to develop management programs for medical equipment. These include protocols and procedures for. Testing to AS/NZS is neither relevant nor necessary in aged care facilities unless the equipment is part of a facility that is intended to be. in AS/NZS In-service safety inspection and testing of electrical equipment. AS/NZS Management programs for medical devices provides.
|Published (Last):||13 April 2011|
|PDF File Size:||15.88 Mb|
|ePub File Size:||17.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
It is designed to cover every economic sector and virtually every activity of the humankind where technical standards may be used. Testing of medical devices is normally undertaken by a professional biomedical technician. Medical devices are a specialist area which is subject to particular standards and legislation. Despite anything in these regulations, the following installations, fittings, and appliances are deemed to be electrically safe if they comply with the ns indicated and are used in accordance with those standards: International Classification for Standards ICS is an international classification system for technical standards.
AS/NZS | On Call Maintenance
Sustaining business, sustaining life. These include protocols and procedures for procurement, acceptance, maintenance activities throughout the service life of the medical equipment, and finally disposal of medical equipment.
All referenced documents are displayed below. Expand the box to see details. What is the Standard about?
Standard & Legislation
The person must test the appliance and either Citations This document is cited by New Zealand legislation. One of the key purposes of this act is to provide a balanced framework to ensure the health and safety of workers and workplaces by “protecting workers and other persons against harm to their health by eliminating or minimising risks arising from work or from potential high risk plant”. It is all about risk management, health and safety and ultimately “peace of mind”.
This document may make references to other documents.
Outlines procedures required to develop management programs for medical equipment. Ensuring you get the right Service Technician is critical to ensure correct compliance. Risk also exists where not all required electrical and performance tests have been completed.
Schedule 2 Citation Context: References This zs may make references to other documents. The technician will maintain the full suite of testing equipment to ensure that all necessary tests are undertaken for both electrical safety and performance checking that the devices work correctly.
Parameters are determined at time of device acceptance in accordance with manufacturers’ specification. A used appliance that is sold or offered for sale is deemed to be electrically safe if, at the time it qs sold or offered for sale, Management of stock General requirements for basic safety and essential performance ISO Medical devices – Application of risk management to medical devices AS ISO Medical devices—Quality management systems – Requirements for regulatory purposes.
The testing they do is normally suitable for fridges and microwaves but not for medical equipment.
Electricity Safety Regulations Ministry: Follow these links to find other Standards similar to the one you are viewing. Physical Inspection Check for signs of damage Check that the device is clean. Correct performance verification is critical to ensure safe use of the medical device for the patient and the operator.
Click Buy for Online Library subscription. Other people who purchased this product also bought Physical Inspection Check for signs of damage Check that the device is clean B.